BioDoc-It2 with its streamlined all-in-one design enables researchers to easily capture high resolution, publication-quality gel images with the touch of the. BioDoc-It and VisiDoc-It Imaging Systems. 2. System Introduction. BioDoc-It™ Imaging System. The BioDoc-It Imaging System enables simple documentation of . A review of the BioDoc-It with Fluor Cam from UVP. Unbiased reviews by scientists available at
|Published (Last):||16 July 2007|
|PDF File Size:||10.33 Mb|
|ePub File Size:||3.90 Mb|
|Price:||Free* [*Free Regsitration Required]|
Installing the Printer Software. Click the down arrow to select the file format. Touch the OK button to save all preferences and go back to the main screen, or the Cancel button to go back to the main screen without saving changes. Carefully biodoc-ti instructions before attempting to install this product.
If one is not plugged in, the TS software will automatically prompt the user for a location. Choose the following settings: Restart the BioDoc-it system. The filter tray is located under the camera bracket on the top right side of the darkroom. Hewlett-Packard Biidoc-it makes no warranty of any kind with regard to More information. Optionally you can convert to standard checkout mode by choosing a payment type.
Digital Photo Picture Frame. Decrease or increase the time by touching the minus or plus button respectively. If the auto save function is enabled, the image will automatically save when the SNAP button is pressed. The developed all-in-one design enables researchers to easily capture high resolution, publication-quality gel images with the touch of the screen.
Quick Start Guide 4. Tilt the screen biovoc-it the desired angle. ArtixScan XL lower base 2. Readjust the Aperture adjustment ring if needed. For complete warranty details and a list of our worldwide offices. HP Scanjet G series.
BioDoc-It Imaging System – PDF
Select a folder or make a new folder. Bio-imagingInstrumentation Life Sciences.
We work around the clock and around biodoc-ig globe to ensure that our products maintain More information. To power down the system, push the Main Power button.
The ST Microscope is both versatile and easy to use with. Adjust the exposure time by first selecting the time interval.
Adjust exposure time 2. Plug the camera cable into the top of the camera and the other end into a USB port on back of the darkroom. Press bioxoc-it Open button to open the lid of biodkc-it printer and load the paper by allowing the loose end of the roll to come off the top of the roll. Status bar modes 8. Multiple slate users can write at. Then, minimize the TS software screen to access the Windows Start button and network setup functions by touching the Windows logo button on the lower-right corner of the TS software.
Ensure that the text on the edge of the filter is positioned so it is right side up when facing the installer. Touch on the corresponding bars, Biodoc-ut, Seconds, milliseconds, from left to right, respectively.
UVP® BioDoc-It™ 220 UV Imaging Systems
Please read carefully and follow all bidoc-it and instructions in More information. Skip to main content. User Guide Wall-Mounting your HP computer User Guide The only warranties for Hewlett-Packard products and services are set forth in the express statements accompanying such products and services.
Transilluminator Will Not Turn On 1. The name of the attached camera will be shown.